eSeasongear Storage Solution for pH & ORP Electrodes/Probes/Sensors, Small Size, 20ml
(HI 70300, HI70300, HI70300S, MA9015 MA9015S)
Milwaukee Instruments MA9025 ORP Probe/Electrode Conditioning Kit, 2x2Oz, MA 9025
*** The MA9025 ORP Probe Oxidation Kit is Discontinued.
*** Per Milwaukee Instruments Tech Support (252-443-3630), end user can use distilled vinegar to oxidize the electrode but it takes 2 to 3 days with the electrode in a jar of 1” to 2” of vinegar to be fully conditioned.
Suggested Solutions for pH Meters / Controllers
1. MA9015 storage solution for pH probe storage.
2. MA9016 cleaning solution for pH probe cleaning.
3. MA9007 pH 7 buffer for 1-point pH calibration.
4. MA9004 pH 4 and MA9010 pH 10 for 2-point pH calibration.
After each use, always rinse the probe with water, clean it with MA9016 cleaning solution and store the probe with MA9015 storage solution.
For probe cleaning - Soak it in MA9016 cleaning solution for about 3 minutes once a day or at least once a week to prevent clogging and to maintain accuracy.
For probe storage - Place a sponge or tissue in the bottom of the cap, and pour some MA9015 storage solution into the cap.
Never let probe dry after conditioning and Never use distilled/deionized water for storage purposes.
Always rinse the pH electrode with water between calibrations and testings.
Check out the "pH Electrode Care" for more details.